Here is a current source you can build for resistance measurement. When the current is held constant, you know as per Ohm's Law the Voltage across Resistor is proportional to Resistance value.

Resistance measurement with Current Source

The supply is +12 and -12, The total voltage across R6 + R7 is 24V. Then 24V / 120K = 0.2mA. The voltage across R6 is (10K * 0.2mA) = 2V. The same is reflected across R5 in this feedback configuration. That means Q3 is a 2V / 1K = 2mA source. If my calculations are right.

Design of a Constant Current Source

There are sources of errors in this circuit. The temperature variation of all resistor values, which is 100ppm for general calculations in 1% MFR. Let us assume you use OP07 which is close to an ideal opamp, but for this application it is not needed. The second error is Ib, the base current of Q3 which may be 0.2mA / Hfe(200) = ~ 1 uA. Then the variation of Hfe, Vcc and Vdd w.r.t. Temperature, should not be overlooked. Use LM7812 and LM7912.

So you see, design knowing that all these components are not ideal. Leakage currents, Humidity, EMI, Stray Capacitance and Inductance and much more. It is just like, even when the motor is fixed firmly on the machine, some parts Vibrate and create a Noise due to Mechanical Resonance. So Build and evaluate your design in the real environment, to learn.

Discover how resistors are color coded - Interactive Java Resistors Tutorial.

## Saturday, November 19, 2016

## Circuits by Application

Analog Circuits - Battery Level Indicator
- Simple Sample and Hold
- Sample and Hold Standby
- Voltmeter Attenuator
- Precision Current Source
- Opamp Supply Virtual Ground
SCR and Triac - Solid State Relay
- Normally Closed AC SSR
- AC-AC-SSR
- DC-DC SSR
- 2N2646 based Pulser
- Drive SCR thyristor
Mains Power | Digital Circuits - Simple Digital Counter
- Running Lights
- Frequency Divider
- Crystal Oscillator
- Simple High speed switch
- Differential TTL converter
Measureall DMM Mixed Circuits | 555 Circuits Discrete Circuits |

Opamp Circuits | Power Electronics - Dual Power Supply
- Single Power Supply
- Battery Backup Supply
- 5V 1A Supply LM2575
- 5V Power Supply L296
- Dual Power Supply
- Tubelight Electronic Choke
- Voltage Doublers Multipliers
- White LED Lamp on Ni-Cd
uC and uP | Instrumentation Circuits Process Control |

## Layers

555-Circuits
555-Circuits-2
555-Circuits-3
80C31-SBC
Analog-Mux
Battery-Circuits
Calibration
Computer-Interface
Digital-Circuits
Digital-Circuits-2
Discrete-Circuits
Discrete-Circuits-2
Discrete-Circuits-3
Discrete-Circuits-4
Embedded-Systems
Fluorescent-Lamp
HV-Circuits
LED-Circuits
LED-Lighting
LED-Meters-Bargraph
Mains-Power
Mains-Power-2
Mains-Power-3
MCS48
Microcontroller-Circuits
Mixed-Circuits
Mixed-Circuits-2
Mixed-Circuits-3
Monostable-Multivibrator
Multimeter-DMM
Neon-Circuits
Opamp-Circuits
Opamp-Circuits-2
Opamp-Circuits-3
Opamp-Circuits-4
Opamp-Circuits-5
Opamp-Digital-Control
Optical-Infrared-Circuits
Oscillators
Power-Supplies
Power-Supplies-2
Process-Control
SCR-Trigger-Circuits
Solid-State-Relay
Switching-Regulators
Temperature-Measurement
Test-Measurement
Test-Measurement-2
Test-Measurement-3
Timers-Counters
Voltage-Multipliers